BSQ-002 single-channel transmetter
|Protection grade||IP65-IP67||Air plug IP65, Waterproof plug IP67 (optional)|
Zero point Zero point
The right output terminal connection is generally connected, red 24V+, green signal +, and black GND. The left side is connected to the input of the load cell and can be docked by color.
After the transmitter and sensor are connected, the zero deviation transmitter can be adjusted:
Open the transmitter box cover, you can see there are two parallel gray potentiometer on the circuit board, by adjusting the upper row 5K potentiometer. The zero point of the transmitter can be corrected. The lower row of potentiometers is used to adjust the fullness, and the user must not make adjustments at will, otherwise the circuit will be disordered.